Rife Health resonator models and their repairs and upgrades.
Rife Health still repair, service and upgrades all its instruments since it imported the first one from the USA in 1999. So far Rife Health was even able to repair "Rife" instruments from other manufacturers as well to ensure that the user has a Rife instrument available even if it is from a competitor. If you experience any technical difficulty please use this guide below to identify the model that you experience difficulty with and complete and submit the online form below:
1. Identify your device:
The Product range since 1999:
The original Rife Wave Generator (no 1) on the left. It was replaced with the slightly wider white Rife Wave Resonator (No 2) on the right in 2003 that has a built-in battery.
The Wave Resonator on the left (no 2) and its larger version the Wave Practitioner (No 3) on the right. The Practitioner model (No 3) was developed for clinics in rural areas where electrical power is not always available. The Practitioner has a larger battery, two sets of electrode plugs so that more than one user can be treated simultaneously and it has a operational light so that the operator can from a distance see if the instrument is still busy with a treatment or not.
The Wave Resonator also came with this keypad in 2015 and 2016 before it was replaced with the new Wave (No 12) in June 2016
The BodySync (No 4)
The BodySync Salveo (No 5)
The WaveSync (No 6)
The Wave III 2013 (No 7)
The Wave III 2015 onwards (No 8)
The Wave Resonator Plus (Introduced 2014) No 9
The Wave Compact (No 10)
The Wave Practitioner (Also called Tri~Wave) (No 11)
The (Swiss) Wave (No 12)
2. Identify the technical problem:
"No current is flowing":
1. First check if unit is switched on.
2. Check the power cord between mains and the unit.
The Wave 3 (No7 and 8) use a hi-fi figure 8 cord. The Wave Resonators (No 1) pre 2005 used a 12 volt charger/transformer. The Wave Resonators from 2005 to 2015 used a 15 volt charger but a 12 volt charger will also work.
The new Wave (From 2016 upwards uses a 5 volt charger).
Check if the charger of your unit is working.
3. On the pre 2015 Wave Resonators (See numbers 1, 2, 3 and 9) if the treatment is running and if there is a "feeling" that no current is flowing it may be that the device is set on "dc" instead of "ac".
4. A faulty cable (That runs from the instrument to the electrodes) may be faulty. Ask a hardware store or electrician just to check with a current meter if the cables are working.
How to connect electrode cables.
The electrode cables have had different plugs over the years. They have holes so that one can stack/connect/extend another electrode cable like when you connect a resonator to a Healthpad.
When a plug needs to be replaced ask a electronic shop or even Builder's Warehouse for a banana plug.
On-line repair form
If you want to repair or upgrade your Rife Health instrument use this form below and Rife Heath will contact you with a repair and/or upgrade quote. Please be very clear and specific in the "My Message" spece how you need Rife Health to assist you: